Corrosion of Steel in Concrete
Corrosion is initiated when materials that are harmful to steel, such as CO2 and chloride from de-icing salt, start to penetrate concrete and reach the steel reinforcement. As an electrochemical reaction, electrons migrate from the anodic zone to the cathodic zone, releasing ferrous ions at the anode and hydroxide ions at the cathode. This will eventually lead to a potential difference between the anodic and cathodic areas at the surface of the steel reinforcement. This results in the creation of rust as a by product. Since rust occupies a larger volume than steel, it exerts internal pressure which causes the surrounding concrete to crack and become damaged. These cracks make their way to the surface of the concrete which causes even more CO2 and chloride to penetrate the concrete and speed up the process of corrosion.
The Cost of Concrete Corrosion
Reinforced concrete structures must be tested regularly to detect and prevent corrosion. As these structures get older, the risk of corrosion in reinforcing steel continues to increase. This is especially important since a large number of structures were built using reinforced concrete between the 1950s and 1970s, especially in bridges.
In North America, the average age of the continents’ 607,380 bridges currently in operation is 42 years old, with some of these structures as old as 80 years. Not only that, but one in nine of these bridges is considered structurally deficient and nearing the end of their initial service life (ASCE). Consequently, these structures require more frequent testing and repairs to eliminate damage and slow down the corrosion process.
Since concrete is the most widely used manufactured material globally, the industrialized world faces billions of dollars in concrete testing and repair costs. This is essential in order to keep structures sound and functional. In Switzerland alone, the annual cost of repairs can amount to between $6.6-$26.3 billion CAD. When taking this information into consideration, it is extremely important to accurately assess the condition of reinforced concrete structures in order to decide if repairs are required immediately.
Detecting the Rate of Corrosion
The electrical response of rebar embedded in the concrete can be determined from the surface of concrete using the four-probe Wenner array configuration (as shown in the following figure) by applying a constant AC current between the outer probes and the measurement of voltage between the inner probes.
By sweeping the frequency of the AC current from low frequency to high frequency, the voltage response of the system changes. The variation of the voltage response of the corroding rebar is different from that of the non-corroding rebar. As show in the figure below, the measured voltage when rebar is not corroded increases in the low frequency zone of the spectrum, but it is almost invariable for the corroded rebar. This basic concept can be applied for the detection of the corroding areas of the reinforced concrete structures from the surface without the need to have an electrical connection to the rebar inside the concrete, unlike other existing non-destructive corrosion measurement devices.
Although the low-frequency impedance response of reinforced concrete can be correlated to the corrosion state of rebar in concrete, its direct measurement is very time-consuming and vulnerable to noise interruption; hence, it is not practical to use it in the field to measure the corrosion rate of rebar inside the concrete.
A Practical Approach to Corrosion Detection
In the CEPRA technology developed by Giatec, the low-frequency behavior of reinforced concrete system is determined by applying a narrow current pulse or a step voltage/current for a short period of time (in a couple of seconds) and simultaneously recording the voltage of the system with a high sampling rate. Using the recorded voltage and the applied current the low-frequency impedance response of rebar in concrete can be extracted, which can be used to determine the state of corrosion in reinforced concrete structures. This patented technology is called Connection-less Electrical Pulse Response Analysis (CEPRA).
The CEPRA technology is implemented in a portable wireless device called Giatec iCOR®. This novel NDT tool employs a complex electrical circuit model for predicting different properties of concrete materials, and steel reinforcement. This electrical circuit is schematically illustrated as shown below. An advanced mathematical algorithm is implemented in the core software of the device. This software processor is responsible for the analysis of certain characteristics of reinforced concrete structure such as the polarization resistance of embedded reinforcement and the ‘real’ electrical resistivity of concrete.
Knowing the polarization resistance, one can easily calculate the corrosion rate from the Stern-Geary equation.
CEPRA vs. Other NDT Devices for Corrosion Detection
The CEPRA technology makes it possible to detect corrosion and measure corrosion rate in concrete from the surface without the need to have a rebar connection. This is great advantage as it allows to perform onsite condition assessment quickly and efficiently. The CPERA technology offers an accurate method of corrosion rate measurement as it confines the polarization area and does not get affected by the surrounding environment.
Unique Features of CEPRA:
- Directional measurement of corrosion rate, concrete electrical resistivity, and electrical resistance of concrete cover
- Fast measurement within seconds
- No need to have electrical connection to reinforcement
The sophisticated technique and the advanced algorithms behind the CEPRA technology are embedded in a hand-held device called iCOR. This device measures the electrical response of the reinforcement inside the concrete without a physical connection to the rebar. As a non-destructive testing tool, the iCOR is an award-winning device that has been recognized for its positive impact in corrosion monitoring and mitigation. Using our patented Connectionless Electrical Pulse Response Analysis (CEPRA) technology, the device is unique in its ability to perform three-in-one concrete testing measurements of: rebar corrosion rate, half-cell potential, and in-situ electrical resistivity. These measurements are critical to the success of rehabilitation projects and to the repair of concrete structures.
For this reason, the iCOR is the most convenient corrosion rate measurement device in the field and offers an innovative research tool for laboratory studies as well. It gives engineers a comprehensive understanding of concrete quality and the level of corrosion; ultimately, it allows them to make faster, more informed decisions when it comes to rehabilitation and repair.
Using the communication tools in iCOR’s Android app, the data, reports and corrosion maps can be easily shared. Here is a quick walk-through video on the iCOR device setup and field testing:
Editor’s Note: This post was originally published in July 2016 and has been updated for accuracy and comprehensiveness.