Giatec RCON2™ is a non-destructive device for measuring the electrical resistivity of concrete specimens in the laboratory without any additional sample preparation requirements. This measurement can easily be made on the same concrete samples that are currently used for the compressive strength testing of concrete.
RCON2™ is fast (measurement time is less than 5 seconds), accurate (utilizing variable frequency method) and flexible (the measurement can be taken with different settings for verification). RCON2™ also allows for continuous measurement of electrical resistivity over time, which can be used to monitor several other parameters such as cracking, moisture transfer and time of setting in concrete specimens.
In concrete materials, the electrical resistivity is correlated well with important durability parameters such as permeability, diffusivity and in general the micro-structure characteristics of concrete.
RCON2™ is a unique device for investigating the micro-structural properties of concrete including:
Giatec RCON2™ provides a reliable method that utilizes electrical resistivity measurement for the quality control of concrete based on the required durability performance. This measurement can easily be made on the same concrete samples that are currently used for the compressive strength testing of concrete. RCON2™ employs an AC impedance technique for the accurate and fast readings that can be continuously obtained using its customizable and user-friendly operating software for various concrete materials.
The electrical resistivity of concrete can simply be related to their pore network characteristics such as pore size and their connectivity, moisture content in the pores and pore solution chemistry. In concrete materials, the electrical resistivity has been well correlated with important durability parameters such as permeability and diffusivity. In addition, this non-destructive test can easily be conducted on fresh or hardened concrete specimens at different ages or various stages of hydration in order to study workability, setting and durability performance of concrete.
The electrical resistivity method has also been applied to investigate corrosion of rebar in concrete, creep, aggregate segregation and freeze and thaw of concrete since they affect the pore network properties. The concrete electrical resistivity techniques is also a suitable replacement for the Rapid Chloride Permeability Test of concrete (as per ASTM C1202) since there is a strong correlation between the electrical resistivity and durability performance of concrete.
|Reading Range||Frequency spectrum||Phase measurement||Impedance accuracy||Phase accuracy|
|1 ~100 Ω||1Hz ~ 30KHz||0 ~ 180°||± 2% ± 2digit||5 % ± 3digit|
|1 ~10 KΩ|
|10 ~ 100 KΩ|
|100 KΩ ~1 MΩ||1Hz ~ 10KHz|
|Frequency||Sampling time||Reading time (minimum)|
|1 Hz ~ 4 Hz||5 seconds||10 seconds|
|5 Hz ~ 30 KHz||1 second||2 seconds|
|Operating temperature||15 ~ 45 °C|
|Operating humidity||30 ~ 80%|
|Storage temperature||0 ~ 60°C|
|Operating voltage/current||100-250 V ± 10%, 60Hz|
|Dimensions of RCON2™||200 x 230 x 70 mm|
|900083||RCON2™ - Full Package-115/230V||RCON2™ unit, Power adapter, Test cable set, Alligator test clip, Sample holder, Verification kit, Fresh concrete probe, User manual, Communication PC software, USB cable, 2 bottles of conductive gels, 2 pairs of contact sponge|
|900082||RCON2™ Device-115/230V||RCON2™ unit, Power adapter, Test cables, Verification kit, User manual, Communication PC software, USB cable|
|900018||Test Cable Set||A pair of test cable for connecting the device to the sample holder|
|900065||Verificaiton Kit||Required to verify the performance of the device|
|900013||Sample Holder - Bulk||Sample holder, 2 pairs of contact sponge|
|900014||Contact Sponge D150 - Pair||150 mm Dia. Cellulose Sponge|
|900015||Conductive Gel - Low Viscosity||250 ml bottle|
|900016||Conductive Gel - Medium Viscosity||250 ml bottle|
|900084||Fresh Concrete Probe||Probe for testing electrical resistivity of fresh concrete|
|900069||RCON2™ Fresh Concrete Test Rod Set||Stainless steel rods|
|Chloride Penetration||High||Moderate||Low||Very Low||Negligible|
|56-Day Rapid Chloride Permeability Charge Passed as per ASTM C1202 (Coulombs)||>4,000||2,000-4,000||1,000-2,000||100-1,000||<100|
|28-Day Bulk Electrical Resistivity of Saturated Concrete (kΩ.cm)||<5||5-10||10-20||20-200||>200|
Submit your review
The efficiency of RCON allows to test a large number of specimens in a very short period of time. Giatec's technical team provides very efficient and timely support.
The RCON device is an amazing tool for the concrete arena. We have successfully used the RCON on a wide variety of projects. It's ease of use coupled with dating relating to permeability helps us determine the appropriateness of concrete mixtures and the impact of the constituents.
Giatec makes some great products. Giatec customer sensitivity and attentiveness also sets the bar for the concrete and construction industries.
I have and will continue to reccomend the RCON device and Giatec to my customers and contacts in the concrete and construction industry.