Giatec RCON™ is a non-destructive concrete testing device for measuring the electrical resistivity of concrete specimens in the laboratory without any additional sample preparation requirements. This measurement can easily be made on the same concrete samples that are currently used for the compressive strength testing of concrete.
RCON™ is fast (measurement time is less than 5 seconds), accurate (utilizing variable frequency method) and flexible (the measurement can be taken with different settings for verification). RCON™ also allows for continuous measurement of electrical resistivity over time, which can be used to monitor several other parameters such as cracking, moisture transfer and time of setting in concrete specimens.
In concrete materials, the electrical resistivity is correlated well with important durability parameters such as permeability, diffusivity and in general the micro-structure characteristics of concrete.
RCON™ is a unique device for investigating the micro-structural properties of concrete including:
Giatec RCON™ provides a reliable method that utilizes electrical resistivity measurement for the quality control of concrete based on the required durability performance. This measurement can easily be made on the same concrete samples that are currently used for the compressive strength testing of concrete. RCON™ employs an AC impedance technique for the accurate and fast readings that can be continuously obtained using its customizable and user-friendly operating software for various concrete materials.
The electrical resistivity of concrete can simply be related to their pore network characteristics such as pore size and their connectivity, moisture content in the pores and pore solution chemistry. In concrete materials, the electrical resistivity has been well correlated with important durability parameters such as permeability and diffusivity. In addition, this non-destructive test can easily be conducted on fresh or hardened concrete specimens at different ages or various stages of hydration in order to study workability, setting and durability performance of concrete.
The electrical resistivity method has also been applied to investigate corrosion of rebar in concrete, creep, aggregate segregation and freeze and thaw of concrete since they affect the pore network properties. The concrete electrical resistivity techniques is also a suitable replacement for the Rapid Chloride Permeability Test of concrete (as per ASTM C1202) since there is a strong correlation between the electrical resistivity and durability performance of concrete.
|Reading Range||Frequency spectrum||Phase measurement||Impedance accuracy||Phase accuracy|
|1 ~100 Ω||1Hz ~ 30KHz||0 ~ 180°||± 2% ± 2digit||5 % ± 3digit|
|1 ~10 KΩ|
|10 ~ 100 KΩ|
|100 KΩ ~1 MΩ||1Hz ~ 10KHz|
|Frequency||Sampling time||Reading time (minimum)|
|1 Hz ~ 4 Hz||5 seconds||10 seconds|
|5 Hz ~ 30 KHz||1 second||2 seconds|
|Operating temperature||15 ~ 45 °C|
|Operating humidity||30 ~ 80%|
|Storage temperature||0 ~ 60°C|
|Operating voltage/current||100-250 V ± 10%, 60Hz|
|Dimensions of RCON™||200 x 230 x 70 mm|
|900083||RCON™ - Full Package-115/230V||RCON™ unit, Power adapter, Test cable set, Alligator test clip, Sample holder, Verification kit, Fresh concrete probe, User manual, Communication PC software, USB cable, 2 bottles of conductive gels, 2 pairs of contact sponge|
|900082||RCON™ Device-115/230V||RCON™ unit, Power adapter, Test cables, Verification kit, User manual, Communication PC software, USB cable|
The following replacement parts are available upon request:
|900018||Test Cable Set||A pair of test cable for connecting the device to the sample holder|
|900065||Verificaiton Kit||Required to verify the performance of the device|
|900013||Sample Holder - Bulk||Sample holder, 2 pairs of contact sponge|
|900014||Contact Sponge D150 - Pair||150 mm Dia. Cellulose Sponge|
|900015||Conductive Gel - Low Viscosity||250 ml bottle|
|900016||Conductive Gel - Medium Viscosity||250 ml bottle|
|900084||Fresh Concrete Probe||Probe for testing electrical resistivity of fresh concrete|
|900069||RCON™ Fresh Concrete Test Rod Set||Stainless steel rods|
Q2: Your website mentions the correlation between ASTM C1202 and the RCON™. Do you have any data comparing the results of the 2 methods?
A: Yes. There is a correlation between the results of the rapid chloride permeability test and those obtained from the electrical resistivity method. Please refer to the following table for the comparison of these two techniques:
|Chloride Penetration||High||Moderate||Low||Very Low||Negligible|
|56-Day Rapid Chloride Permeability Charge Passed as per ASTM C1202 (Coulombs)||>4,000||2,000-4,000||1,000-2,000||100-1,000||<100|
|28-Day Bulk Electrical Resistivity of Saturated Concrete (kΩ.cm)||<5||5-10||10-20||20-200||>200|
Q3: What is the measuring range of RCON™? What is the maximum (kΩcm) that can be measured?
A: The measurement range for the RCON™ is from 1ohm to 1Mohm. For a cylindrical sample of 10 x 20 cm the maximum measurement is ~4,000 kΩcm. This range can be increased to ~40,000 kΩcm using a high-resistivity accessory provided by Giatec.
Q4: What kinds of requirements are needed to run the software?
A: The requirements needed to run the software are a PC computer with minimum Pentium 4 Processor, Windows XP/Windows 7/Vista operating system and at least 500 MB Ram.
Q5: What are the standards that were used to develop the equipment?
A: RCON™ was developed based on an upcoming ASTM standard which will be published soon. There is a similar standard in AASHTO for the surface electrical resistivity measurement, but RCON™ device measures the bulk electrical resistivity which provides a more accurate result.
Q6: Does this device measure compressive strength?
A: No, it does not. RCON™ can easily and quickly measure the electrical resistivity of concrete. This parameter is closely correlated with the setting time of fresh concrete and the durability characteristics of hardened concrete.
Q7: Does this device measure the sustainability of concrete and the ingress of salts and chlorides?
A: RCON™ measures the electrical resistivity of concrete. This parameter can be used to estimate the durability and sustainability of concrete in a specific exposure environment. Giatec offers another device called Perma™ which measures the rapid chloride permeability of concrete based on the ASTM C1202. For more information on this device, please click here.
Q8: In some countries, the operating voltage is 210-220 V and the cycle is 50 Hz. Is RCON™ compatible with these specifications? Would this difference of cycles affect the test results?
A: The power supply of the RCON™ supports an input voltage in range of 100-250 V with the frequency range of 50-60Hz. The results wouldn’t be affected at all by these variations.
Q9: Is the item D00101A-11 Fresh concrete probe mandatory for performing tests? Will it be useful in supplementing our results?
A: The fresh concrete probe is an optional item; it is only needed for studying the setting of concrete. If you are not doing this type of testing, you do not need this item.
Q10: How do you measure the change in water content and time of setting in concrete specimen?
A: Using the special fresh concrete probe designed to be embedded in concrete, you can measure the water content and the setting time by monitoring the electrical resistivity versus time. The device comes with software that can log the data. However, calibration is required to correlate the change of the electrical resistivity to the actual water content and the setting time of concrete.
Q11: Are the ASTM & AASHTO Standards still under development?
A: AASHTO standard is already published. You can obtain more information from the AASHTO TP 95-11 document. The ASTM standard which is based on the bulk electrical resistivity of concrete is currently under development and will be published soon.
Q12: Is the concrete specimen used in this test cylinder or can we use a cube sample size (150mmx150mmx150mm)?
A: You can fit the cylindrical samples with the maximum size of 150mmx300mm and the cubic sample with the maximum size of 100mmx100mm in the regular sample holder that comes with the RCON™. However, if you need to measure a 150mmx150mm cubic sample, we can customize the sample holder fixture in order to accommodate this size.
Q13: How many samples can be tested simultaneously?
A: Only one sample can be tested at a time with the current test set-up that we have. However, the test is very fast, taking only 3-5 seconds to accurately process, allowing for many samples to be tested in a short period of time.
Q14: What is the list of users of this product?
A: Our user list includes several North American, European, Australian and Middle Eastern universities and concrete producers. Among these are universities such as University of Michigan, Carleton University, University of Ottawa, University of Toronto, Purdue University, and United Arab Emirates University.
Q15: Do you have a representative or supplier outside North America?
A: Currently we do not have suppliers outside North America. We manufacture and ship the devices directly from Canada.
Q16: How will the product be taken care of in terms of warranty services?
A: Giatec takes care of any service and calibration cost incurred during the warranty period and will provide online unlimited technical support 24/7 to our customers after the warranty period, free of charge.
Q17: In terms of sensitivity, accuracy and calibration, can the RCON unit be equated or correlated in any way to any chloride diffusion testing of concrete?
A: RCON™ is fully calibrated before shipping and, moreover, comes with a verification kit for the customer to verify the test results from time to time if deemed necessary. The device has been designed to provide high sensitivity and accuracy required for concrete research. CLICK HERE for the RCON™ user manual that contains more information regarding the accuracy and sensitivity of RCON™ as well as its calibration and verification procedure.
Q18: What is Giatec’s procurement/sales procedure?
A: Step 1: We provide a quotation of the device and its accessories to the customer. Step 2: customer sends us a purchase order or an email confirming the prices listed in the quotation. Step 3: We process an invoice in reference to the purchase order and include shipping, handling, and insurance costs. Step 4: Once we receive the payment from customer (either by wire transfer or a cheque in advance) we deliver the product which will take 1-2 weeks (depending on location) from date of payment and include a receipt for your records. Step 5: Follow-up with the customer and assure that the product was delivered successfully and everything is in good condition.
Q19: How to use RCON for conductivity measurement for fresh concrete ?
A: The conductivity of fresh concrete (which is the inverse of its electrical resistivity) can be measured using the fresh concrete probe provided by Giatec or using a similar setup. This fixture incorporates two stainless steel rods (with insulating parts at the ends) embedded in fresh concrete. The rods are connected to the RCON device for monitoring the increase in the electrical resistance between them as the concrete sets and hardens. The measured electrical resistance can be converted to electrical resistivity knowing the conversion factor which depends on the geometry of the test container. This conversion factor can be easily obtained by measuring the electrical resistance of a salt solution (with known conductivity) in the test container.
The efficiency of RCON allows to test a large number of specimens in a very short period of time. Giatec's technical team provides very efficient and timely support.
The RCON device is an amazing tool for the concrete arena. We have successfully used the RCON on a wide variety of projects. It's ease of use coupled with dating relating to permeability helps us determine the appropriateness of concrete mixtures and the impact of the constituents.
Giatec makes some great products. Giatec customer sensitivity and attentiveness also sets the bar for the concrete and construction industries.
I have and will continue to reccomend the RCON device and Giatec to my customers and contacts in the concrete and construction industry.
Giatec is driving breakthrough innovations in the concrete industry, bringing real-time data on concrete performance to the forefront of every construction jobsite. Their suite of hardware and software products, including non-destructive testing technologies, IoT solutions, and wireless sensors, analyze data throughout concrete’s lifecycle, helping to build structures faster, safer, and more economically.
Combining wireless concrete sensors and mobile apps, Giatec has developed smart IoT-based technologies, like SmartRock™ concrete maturity sensors, for real-time monitoring of concrete properties. This provides critical information to contractors so they can make efficient and informed decisions on the jobsite, ultimately saving time and money. Ready-mix producers also leverage the SmartRock™ Plus cloud, which gives them access to concrete performance data to optimize their mix designs. Learn More about Giatec's Suite of Products.